The AVQ1020 is feature rich, including alarm logging and reports.
We were looking for a spectral monitor for our transmitter site. In particular, we required a product that would allow our master control operators to remotely monitor the RF performance of the DTV transmitter.
I preferred an analyzer that would simply transfer RF measurements by way of its embedded webpage vs a remote desktop client, mainly for IT security purposes. The second major requirement was simple operation as not all master control operators are deeply technical by training.
What we found was the Avateq AVQ1020. This product complemented our transmitter monitoring tools comprised of web based remote control system and transmitter webpage perfectly. The Avateq provides exact RF data to help trouble shoot system performance from our studio facility’s master control room.
Figure 1. Basic measurements are on the Signal Statistical page.
We connected the Avateq to our roof-top antenna, in order to verify over-the-air performance. We have another RF analyzer that was purchased with the transmitter and connected directly to the mask filter. However this product is 5 or 6 years old and does not have any remote access. The first task was to verify the monitoring information provided by the AVQ1020 to ensure it correlated to the older analyzer. Sure enough the measurements were exact or within ≤ 0.1%, not having the older analyzer calibrated in a few years; we were uncertain of the difference.
Figure 2. The Spectral display.
Now knowing that the remote monitoring tool correlated to our reference, we connected to the network. The basic measurements of SNR, Shoulder attenuation, Frequency offset and much more are presented in a simple to understand Signal Statistical page (Figure 1). This is perfect for mastcon operators. During periods of inclement weather (specifically winter) we have an immediate indication of any degradation in our signal, we can now be pro-active.
Being able to see the Spectral and Constellation display remotely is a true asset of this machine (Figures 2 & 3). The flexibility provided by the AVQ1020 allows my engineering and master control teams to be on top of any problem, and proactively; for example monitoring degradation over time.
Figure 3. The Constellation display.
The AVQ1020 is feature rich, including alarm logging and reports; alarms are completely user configurable. Other displays of this powerful monitoring tool include CCDF, Eye Diagram, Echo Profile, SFN Drift, Am-Am & Am-Pm, LUT, Group Delay, LP Filter, and Channel Response. Its a lot of power from a web based monitoring product that exactly meets all of our requirements and then some.
You might also like...
In part 8 of the series “Data transmission and storage”, consultant John Watkinson looks at some of the intricacies of RF transmission.
Saving dollars is one of the reasons broadcasters are moving to IP. Network speeds have now reached a level where real-time video and audio distribution is a realistic option. Taking this technology to another level, Rohde and Schwarz demonstrate in…
Two years ago proponents of ATSC 3.0 began looking for a facility to host the testing of the country’s next-generation TV broadcasting standard. Similar to the early days of digital broadcasting—when the 8-VSB modulation scheme was being considered for use…
The Duchess of Windsor, the American wife of the man who was previously England’s King Edward VIII, famously said: “You can never be too rich or too thin”. If she’d lived in the 21st century, she might have added: …
Images from the past can trigger nostalgic memories. Recently, I saw an old film that featured a Heathkit, an assemble-it-yourself table radio with majestic glowing tubes. It brought back memories of my childhood days as an amateur “kitchen table” engineer wh…